Material Information All the cuvettes you request are consequently sent in optically coordinated with sets. In the event that you need to buy another cuvette to coordinate with a more established one, if it’s not too much trouble, let us know. Additionally with respect to our materials, the accompanying data is accessible:
● Black quartz information
● Chemical compatibility
● Thermal properties
● Fluorescence, refractive index, and other optical qualities
Light Path, Tolerances, and Other Specifications Regarding general details, kindly note that each part number has its own resistances. We can likewise give data in regards to window levelness, clean, and parallelism. If it’s not too much trouble, get in touch with us for this, just as extra material information recorded previously.
Frequently Asked Questions
Q: I have a specific instrument, and I need to make sure the cuvettes in this catalog are compatible…
A: The cuvettes in this inventory are ensured to work in a heap of various instruments. Call or email to demand a change diagram between the part quantities of your instruments’ cuvettes and our cuvettes.
Q: What is the Z-dimension (Zd) or Center Height?
A: The Z-measurement is the range from the lower part of the cuvette to the focal point of its example chamber. In situations where the cuvette’s cleaned window is little (where normally the whole cuvette is dark beside the opening window), the Z-measurement of the cuvette ought to be equivalent to the stature of the instrument’s light shaft. The Z-measurement is generally 8.5 or 15mm, contingent upon your instrument’s maker. Just certain cuvettes rely upon the right Z-measurement to be utilized. These cuvettes have Z-measurement (Zd) data gave in their information table.
Cleaning and Handling Cuvettes should be circumspectly spotless. Optical surfaces ought not be contacted as oil smircesh are hard to eliminate. Immaculateness of solvents utilized for test and cleaning is significant. Just refined water or spectrophotometric grade solvents ought to be utilized to maintain a strategic distance from tainting from sullied solvents. Straightaway after use, cuvettes ought to be washed and absorbed refined water. For cleaning, utilize a gentle cleanser arrangement containing no lanolin, oils, or insoluble matter. To eliminate difficult impurities, cuvettes can be doused for a couple of moments in gentle sulfuric corrosive. Never permit cuvettes to absorb hot concentrated acids, soluble bases, or different specialists which may scratch the cleaned optical surfaces. In the wake of cleaning with cleanser and additionally corrosive, cuvettes ought to be completely flushed with refined water. Try not to blow the cuvette with air to dry them. It is smarter to speed vanishing by methods for pull. If it’s not too much trouble, see page 33 for cuvette cleaners. If it’s not too much trouble, reach us for the more information.